Hello guys, in this article you will be learning What Is Loading Effect of Electrical Measurement such as ammeter, voltmeter, and a multimeter. So before reading, let’s understand the definition of the Loading Effect.
Loading Effect of Electrical Instruments
What is Loading Effect in Electrical Instruments
The loading effect of electrical instruments is defined as the degree to which the electrical measurements impact the electrical properties of a circuit. These measurements are mainly ammeter, voltmeter, and a multimeter.
Loading Effect in various Electrical Measurements
In electrical measurements mainly three types of instruments included such as ammeter, voltmeter, and a multimeter. So let’s discuss the loading effect of all these three instruments.
Loading Effect of Ammeter
An ammeter is an electrical instrument that measures AC or DC in Amperes. To measure the current flow in a given electrical circuit, the ammeter should be connected in series. The ammeter is designed for small internal resistance (10 – 100 Ω). Now for impact working of an ammeter, there are two possible cases exists:
In this case, the ammeter is connected to a low resistance circuit then the overall resistance of the circuit is dragged to a noticeable amount. So, this means that when an ammeter having a value of 50 ohms internal resistance is connected to a 100 ohms circuit then finally its value is rapidly increased to 3 times. So this finally results in an increase in voltage as well as current by 3 times.
Now on the other hand when the internal resistance of the ammeter is high the same value of impact is observed. And the parameter of an electrical circuit changes or varies.
You may interested: Engineering mechanics textbook free download
Loading Effect of Voltmeter
A voltmeter is also an electrical device, this gives the potential difference between two points of an electrical circuit. A voltmeter is always connected in parallel with the electronic components for measuring the voltage. Voltmeters are designed especially for high internal resistance. It’s valued generally range from (10-20) MΩ. For voltmeter also two possible cases exist:
When we connect a voltmeter to high external resistance it overall value rapidly decreases. Let’s consider an example of connecting the voltmeter to external resistance of 10 MΩ and internal resistance of the circuit to 7.5 MΩ resistors in parallel. Then it is observed that the overall resistance changes to 4.279 MΩ.
In the second case, it’s observed that when the internal resistance is small and connected to high external resistance then it is observed that overall resistance is seen to be changed.
As we have observed in the above both cases change the working of the circuit thus we get appropriate results from the measuring instruments.
Loading Effect of Multimeter
The effect of measuring voltage with a multimeter can be derived as given in the below PDF. To learn about this, you need to download the pdf from the link provided below.
Download PDF :- HERE
All the contents provided in this website are presented specially for educational purpose only. We aim that everyone should be capable of getting freely educated. All the information provided here based upon various book reference and do not claim for any copyright. If still you have any complaint regarding the content before taking any action please inform us so we will remove the content as per your demand. you can contact us here.
After going through all the above points mentioned above. Hope you can understand the Loading effect of electrical measurement instruments. If still, you have any doubt you can reach out in the comments box provided below. We will try to reach you as soon as possible. Thanks a lot for reading the post.